Current Size: 100%
You are hereHome » Events and Media
If above video does not display: https://www.youtube.com/watch?v=f-tva2zj0H0
A general information video on glaucoma using diagrams of eye structure.
- corneal thickness expressed in mmHg rather than micrometers, as Dr. Ritch pointed out.
- Although cornea edema is mentioned, there is insufficient mention of cornea biomechanics as a source of error in Goldmann applanation tonometry.
- The assumptions in the Modified Imbert Ficks law are ignored. For example, the assumption that surface tension and force to bend corneal cancel each other out, is presented without qualification.
- The explanation of how non-contact tonometers compute intraocular pressure is incomplete (and possibly incorrect).
- Propagation of the unsupported idea that Goldmann tonometry is the "most accurate" method in spite of recent advancements such as the Pascal Dynamic Contour Tonometer and the Reichert Ocular Response Analyzer.
- It is understandable that a short video cannot go into every detail, but I would have liked to see a slightly less dogmatic presentation of the same material.
If above video does not display: https://www.youtube.com/watch?v=51MheK88170